DDM-2050L Diameter Laser Gauge Measurement Display
The cable diameter detection device is one kind of laser gauge for the diameter of cable or wire, it’s based on a laser diameter measurement technique, it’s one of the best device of diameter measurer.The DDM2050 comes with DSP-4A Monitor and controller, the DSP-4A monitor offering remote display and PI control signal output.
The Features of DDM series laser diameter gauge.
Very short exposure times
CCD image measuring technique with laser light sources
DSP data processing
Technical Parameters (Specifications)
|Product Diameter||0.1 ~25 mm||0.1 ~48 mm||0.5 ~95 mm||0.1 ~25 mm||0.1 ~48 mm|
|Accuracy||± 2 μm||± 2 μm||± 5 μm||± 2 μm||± 2 μm|
|Exposure Time||200 μS||200 μS||200 μS||200 μS||200 μS|
|Measuring Rate||600 measurements/sec/axis||600 measurements/sec/axis||600 measurements/sec/axis||600 measurements/sec/axis||600 measurements/sec/axis|
|Power Supply||AC220V±10% , 50/60 Hz, 15 VA||AC220V±10% , 50/60 Hz, 15 VA||AC220V±10% , 50/60 Hz, 15 VA||AC220V±10% , 50/60 Hz, 15 VA||AC220V±10% , 50/60 Hz, 15 VA|
|Dimensions||408 x 180 x 82 mm||408 x 180 x 82 mm||504 x 203 x85 mm||355 x 330 x 82 mm||355 x 330 x 82 mm|
Application and Features
DDM series laser diameter gauge is used for online diameter measurement for electric wire and cable, optic fiber cable, tubes, and wires. Typical applications: Measurement of the outer diameter of fine wire, Outer diameter lump, and neck down detection at ultra high speed, Outer diameter measurement in harsh environments.All digital image process technique and dedicated ultra-large-scale integrated circuit, which ensure high stability, compact size, and low power consumption.
RS-485 communication interface which makes the controller to be connected easily to an industrial computer, PLC and touch panel.
DDM series laser diameter gauge can be mounted before or after the cooling trough. When it is mounted before the cooling trough, an optimized controlling effect may be obtained through in-time feedback due to the proximity of the measuring head to the extruder. But at this time the plastic coating is not cured, so the set nominal value must be accordingly. When it is mounted after the cooling trough, the measured value is the real diameter of the wire or cable. But the controlling effect is not so good because of the large control lag. Be sure to blow dry the tested wire or cable, otherwise, the measuring precision will be affected by the water coating around the wire or cable.
In practical using, the nominal value shall be increased appropriately according to the water content of the test object.
Check out the link below for the Application solutions.
DSP-4A Monitor Specifications
|Power supply:||100 ~ 240VAC, 50~60Hz|
|Power consumption:||< 5W|
|Operating temperature:||0～45 ℃|
|Atmospheric humidity:||≤ 90% (free of condensed water)|
|Display format:||5-digit LED digital display|
|Max commun distance:||1000 m|
|Control input:||– 10V ~ +0V, input impedance 50 kΩ|
|Control output:||– 10V ~ +10V, max 10 mA|
|Alarm output:||Relay contact, max load 3A, 250VAC|
|Dimensions:||48mm × 96mm × 112mm|