CDM-28 Non-contact Compact Size Diameter Laser Scanning Gauge
CDM-28 diameter gauge is our company’s economical products series, compact size, lightweight reliable product. Currently, it is used for wire cable, fiber measurement Especially CDM-03 can measure 0,01mm wire Diameter Gauge Features 1. Non-contact laser diameter measurement, high speed, high accuracy, and good repeatability; 2. Real-time outside diameter display improve quality control and production efficiency reduce the waste of materials; 3. Various preset functions nominal value upper and lower tolerance, alarm control presets, easy to operate.
The gauge applications include:
- Optical Fiber
- Glass Fiber
Technical Parameters (Specifications)
|Measuring Range||0.01-2 mm||0.2-25mm|
|Accuracy||±1.0 um||±1.0 um|
|Power Supply||220VAC ±15% 50/60Hz|
|Working Temperature||-10-40 Degree C|
|Com Port||RS485 Modbus/Profibus ( Option )|
|Analogs Output||16 Bit 10VDC|
One basic CDM-28 measurement tool is composed of laser gauge and remote controller, it is able to be connected with different types of remote display.
The remote controller is mainly used for automatic control, which obtains measured data from the measurement gauge. PID signal comes out through the comparison between measured diameter
with a preset diameter and controls traction speed thereby controls wire diameter. In addition, a remote controller has a full-function operation panel and standard Serial I/O which communicates with PC
and PLC and supports diameter monitor software.
Laser gauge equals to a high-precision diameter sensor which not only measures online in real-time but also boosts strong data communication function.
For some occasion requiring only diameter measure, the laser gauge with the operating panel is competent for diameter measurement, display, and over-deviation alarm.
For some occasion not only requiring diameter measure but also automatic control, the solution includes a remote controller and laser gauge, which communicates with the former via serial I/O.
A high-precision laser scanning system and embed digital processing electronics are embedded in the measurement gauge. The signal of diameter processed by scanning system, it is converted to digital signal firstly, and then comes out with real diameter and related data through a series of operation.
The laser beam irradiates to the rotating prism and after reflection and through lens 1, it becomes a parallel beam. The parallel beam converges to the receiving component through lens 2. When placing EUT on the focal plane (FP) of lens 1, part of lights are shadowed, thus becomes electrical signal on receiving component and then comes out diameter and a serial of related signals after processing by measurement gauge.
The communication interface of the measurement gauge contains one serial I/O, it transfers measured data to a peripheral device (e.g. remote controller, PC, and PLC, etc.) through the serial I/O and to remote controls and displays through data channels.